Bottom line: All conditions must be met to be MRI permissible.
Medtronic Synchromed Implantable Infusion Pump
Risks: Device malfunction, Thermal
- Ensure patient can be deprived of medication infusion during the MRI scan.
- The pump MUST NOT BE oriented exactly 90° to the Z axis of the MRI scanner.
- Static magnetic field strength up to and including 1.5T.
- Horizontal bore MRI scanner only.
- Verify pump settings are correct after scan completed.
- The pump can become disabled or altered by the MR field and infuse a different amount of medication than indicated. The pump settings must be verified after the scan.
- The conditions and comments only apply to the Medtronic Synchromed pump models. Prometra Infusion Pump is a different implantable infusion pump with different specifications.
- Also Known As: Implantable infusion pump, drug infusion pump, Baclofen pump, intrathecal pump
Bottom Line: All conditions must be met to be MRI permissible.